Meas
Class 
Measurement Name 
Description 
"Standard" 
"S11",
"S21", and
so forth
"A_1","A_2",
and so forth 
Sparameter
name
Unratioed parameter names with notation: "receiver_source
port"
See Balanced
Sparameter measurement names 
Active Hot Parameters 
"HotS11"
"HotS31"
"HotS13"
"HotS33"
"IPwr"
"OPwr"
"Gamma"
"Pmax"
"Xs(3,3)"
"Xt(3,3)"
"Xf(3,1)"
"DeltaOPwr" 
Learn
about Active Hot parameters 
"Vector
Mixer/Converter" 
"S11"
"VC21"
"S22" 
Learn
about VMC parameters 
"Scalar
Mixer/Converter" 
"S11"
"SC21"
"SC12"
"S22"
"Ipwr"
"RevIPwr"
"Opwr"
"RevOPwr" 
Learn
about SMC parameters 
"Gain
Compression"
Learn
more
"Gain
Compression Converters"
Learn
more 
GCA
and GCX: 
"CompIn21" 
Input
power at the compression point. 
"CompOut21" 
Output
power at the compression point. 
"CompGain21" 
Gain at
the compression point. 
"CompS11" 
Input
Match at the compression point 
"RefS21" 
Linear
Gain 
"DeltaGain21" 
CompGain21
Linear Gain 
"S11",
"S21", "S12", "S22" 
Standard
Sparameters; measured at port 1 and port 2 
GCX
 All Gain Compression
parameters (except S21 and S12) plus the following: 
"S11"
"SC21"
"SC12"
"S22"
"Ipwr"
"RevIPwr"
"Opwr"
"RevOPwr" 
Mixer parameters 
"Noise
Figure Cold Source"
Learn
more
"Noise Figure Converters"
Learn
more 
Noise
Figure AND NFX: 
"NF" 
Noise figure 
"ENR" 
Validate noise source
measurements. 
"TEff" 
Effective noise temperature. 
"DUTRNP"
"DUTRNPI" 
DUT noise power ratio.
(Noise power expressed in Kelvin divided by 290). 
"SYSRNP"
"SYSRNPI" 
System noise power ratio 
"DUTNPD"
"DUTNPDI" 
DUT noise power density.
(Noise power expressed in dBm/Hz). 
"SYSNPD"
"SYSNPDI" 
System noise power density.

"OvrRng"
(Opt 029 Only) 
Indication that the noise
receiver is being over powered. 
"TRcvr"
(Opt
029 Only) 
Temperature reading (in
Kelvin) of the noise receiver board. 
Noise
Figure ONLY  NOT NFX: 
"S11",
"S21", "S12", "S22" 
Standard Sparameters;
measured with the port1 and port2 noise switches set for
noise mode. 
"A_1","A_2"
...and so forth. 
Unratioed
parameters; with notation:
"receiver, source port" 
"GammaOpt" 
Optimum
Complex Reflection Coefficient 
"Rn" 
Noise
Resistance 
"NFMin" 
Minimum noise figure that
occurs at GammaOpt 
NFX
ONLY: 
"S11"
"SC21"
"SC12"
"S22"
"Ipwr"
"RevIPwr"
"Opwr"
"RevOPwr" 
Mixer parameters 
"ALO1","
BLO1" ...and so
forth. 
Test port receiver at
LO1 frequency 
"R1_1",
"B_2" ..and
so forth. 
Unratioed parameters with
notation:
"receiver_source port" 
"Swept IMD"
"Swept IMD Converters"
Learn
more 
There
are over 150 possible Swept IMD parameters, too many to
list here.
Build the parameters with
the Swept
IMD Parameter dialog, then copy the parameter name
to the remote command.
The following are a few
example parameters: 
"PwrMainLo" 
Absolute power of the
Low tone at the DUT output. 
"IM3" 
Power of the third product
relative to the average power of the f1 and f2 tones measured
at the DUT output. 
"OIP3" 
Theoretical power level
at which the third product will be the same power level
as the average of the main tones at the output of the
DUT. 
"IM Spectrum"
Learn
more 
"Output" 
View signals OUT of the
DUT and into PNA port 2 (B receiver). 
"Input" 
View signals IN to the
DUT (R1 receiver). 
"Reflection" 
View signals reflected
off the DUT input and back into PNA port 1 (A receiver) 
"IMx
Spectrum Converters"
Learn
more 
"Output" 
View signals OUT of the
DUT and into PNA port 2 (B receiver) 
"Differential
I/Q"
Learn
more 
Create
custom parameters using DefineParameter Method,
then specify your custom parameter name here.
The following are default parameters: 
"IPwrF1"
"OPwrF1"
"GainF1" 
Input
Power over F1 range
Output Power over F1 range
Gain over F1 range 
"Spectrum
Analyzer"
Learn
more 
"a<n>"
"b<n>"
where
<n> is the
port number to measure
"ImageReject<n>"
where
<n> is the image reject trace 
Reference
receiver
Test port receiver 